Question? Just ask us!
Free Encyclopedia of Building & Environmental Inspection, Testing, Diagnosis, Repair
InspectAPedia ® Home
ENVIRONMENTAL HAZARDS - INSPECT, TEST, REMEDY
AIR FILTERING STRATEGIES
AIR POLLUTANTS, COMMON INDOOR
AIR QUALITY IMPROVEMENT STRATEGIES
AIRBORNE PARTICLE ANALYSIS METHODS
ALLERGEN TESTS for BUILDINGS
ANIMAL or URINE ODOR SOURCE DETECTION
ASBESTOS IDENTIFICATION IN BUILDINGS
ATTORNEYS and EXPERT WITNESSES
BACKDRAFTING HEATING EQUIPMENT
BIBLIOGAPHY for ENVIRONMENTAL HEALTH, MOLD, IAQ
BLACK MOLD, HARMLESS COSMETIC
BLACK MOLD, TOXIC & ALLERGENIC
BOOKSTORE - ENVIRONMENTAL
BUILDING SAFETY HAZARDS GUIDE
Cadmium in the home
CARBON MONOXIDE - CO
Carbon Nanotube Hazards
CARPETING & INDOOR AIR QUALITY
CHEMICAL CONTAMINANTS in WATER
CHINESE DRYWALL HAZARDS DS
COMBUSTION PRODUCTS & IAQ
CPSC Indoor Air Pollution Book Online Copy
DIRECTORY of MOLD / ENVIRONMENTAL EXPERTS
Diethylstilbestrol - DES
DUST SAMPLING PROCEDURE
EMERGENCY RESPONSE, IAQ, GAS, MOLD
EMF ELECTROMAGNETIC FIELDS & HUMAN EXPOSURE
ENVIRO-SCARE - PUBLIC FEAR CYCLES
ENDOCRINE DISRUPTORS at BUILDINGS
FEAR of MOLD - MYCOPHOBIA
FLAME COLOR, BLUE vs YELLOW COMBUSTION
FLOOD DAMAGE ASSESSMENT, SAFETY & CLEANUP
FLOOR TILE ASBESTOS IDENTIFICATION
GAS EXPOSURE EFFECTS, TOXIC
GAS EXPOSURE LIMITS & STANDARDS
HEATING OIL EXPOSURE HAZARDS, LIMITS
HOME HEATING SAFETY
INDOOR AIR QUALITY & HOUSE TIGHTNESS
INDOOR AIR QUALITY IMPROVEMENT GUIDE
LEAD POISONING HAZARDS GUIDE
Legionella Legionnaires' Disease t
LIGHT, GUIDE to FORENSIC USE
GAS LP & NATURAL GAS SAFETY HAZARDS
METHANE GAS SOURCES
MSDS Material Safety Data Sheets
MYCOPHOBIA, STAINS MISTAKEN for MOLD
MYCOTOXIN EFFECTS of MOLD EXPOSURE
ODORS GASES SMELLS, DIAGNOSIS & CURE
OIL, HEATING, EXPOSURE HAZARDS, LIMITS
PARTICLE SIZES & IAQ
Particulates & Allergens Indoors
PESTICIDE EXPOSURE HAZARDS
PET ALLERGENS / PET DANDER
PLASTIC ODORS-SCREENS, SIDING
PLUMBING SYSTEM ODORS
PVC - VINYL BUILDING PRODUCTS
RADON HAZARD TESTS & MITIGATION
SAFETY HAZARDS GUIDE
SAFETY HAZARDS & INSPECTIONS
METHANE GAS HAZARDS
SEWAGE BACKUP TEST & CLEANUP
SICK HOUSE IAQ QUESTIONNAIRE
STAIN DIAGNOSIS on BUILDING EXTERIORS
STAIN DIAGNOSIS on BUILDING INTERIORS
UFFI UREA FORMALDEHYDE FOAM INSULATION
VENTILATION in BUILDINGS
VINYL CHLORIDE HEALTH INFO
VINYL Siding or Window PLASTIC ODORS
VOCs VOLATILE ORGANIC COMPOUNDS
WATER ODORS, CAUSE CURE
WATER TESTS, CONTAMINANTS, TREATMENT
WORLD TRADE CENTER 9-11 DUST PHOTOS
This article explains how to use EMF or ELF measuring instruments when performing electromagnetic field (EMF) or electro-magnetic radiation EMR measurements either by engaging a professional or by consumers using low-cost instruments which measure EMF exposure levels in gauss or milligauss. We discusses sources of error and variation in EMF measurements and we review and make suggestions for using several low-cost EMF measurement devices to determine the instantaneous electromagnetic field exposure.
Green links show where you are. © Copyright 2014 InspectApedia.com, All Rights Reserved.
The information provided here is for research and study purposes. The author makes no representation of unique expertise on this topic, other than having field experience in EMF measurement, having studied technical literature and having conversed with other experts and authors in the field for a number of years.
For position-insensitive equipment, a single reading is usually provided, directly in mG.
For position sensitive equipment you'll find an enormous range of response depending on the angle and direction in which you hold the measurement device.
While this type of instrument is more work to use (see calculations below and in spread sheet) it provides more clear indication of when you're approaching a field. Some instruments do not provide a reading directly in mG and you'll have to simply record the "raw" measurements and to convert them later.
Some readers may be confused between two common terms used to describe the strength of an electromagnetic field: Gauss and Milligauss. Both of these terms measure the same effect, but at different ranges of strength level. It's simply a matter of moving the decimal point, as we demonstrate in our photographs above.
In our first photo (above left) the ELF EMF meter has been set to the Gauss range and we are measuring a field strength of 0.03 Gauss with our meter touching a small power transformer.
In our second photo (above right) the same ELF EMF meter has been set to its more-sensitive Milligauss range and we are measuring 25 milligauss.
Because usually the electromagnetic field strength around residential properties is very weak, most measurements will probably be made in the milligauss range. 0.03 Gauss is the same as 30 Milligauss. Our instrument, more accurate in the lower milligauss range, shows that we are actually measuring 25 milligauss, a number which was rounded up to 0.03 Gauss when the scale was shifted.
Definitions of Gauss and Milligauss for Measuring Electromagnetic Fields - EMF
Gauss is a measurement scale used to measure the strength of an electromagnetic field (EMF). The gauss level measured in a given location depends on the strength of the source of the EMF and the distance from the source. The measured field strength of an EMF falls off quite rapidly with distance (field strength declines as the square of the distance from the source).
When electrical current flows through a wire (such as in an electrical distribution wire or inside of an electric motor) an electromagnetic field is produced. Power lines, local electrical wires, air conditioning motors, computers, TVs, hair dryers, toasters, all devices that use or transport electricity will produce an electromagnetic field. Usually the strength of these fields is low (Electric fan 1 mG, TV 1mG, chain saw 2.5 mg(?) at chest height, electric stove 6 mG, older electric blankets before circuit redesign 20 mG, and under a distribution line for electrical power transmission, 2-20 mg depending on the line height, KVA rating, and actual level of usage at the time of measurement.
If there is a concern for measuring exposure or possible exposure to electromagnetic fields, it's critical that we have a correct understanding of the levels of exposure that are being examined. For example, one of our readers informs us that that manufacturer of his pacemaker recommends avoiding exposure to electromagnetic fields stronger than 5 gauss. (EMF generated from small electric motors (such as a chain saw) measured with the saw held in normal operating position may be as much as 2.6 Gauss at the user's chest.)
When recording EMF measurements be certain that you've got the right order of magnitude.
1 Milligauss (mG) = 10-3 Gauss, or 0.001 Gauss
Published sources provide lots of examples of producing and measuring electromagnetic fields at a known gauss level. For example, using a 1.5V battery and a 150 Ohm resistor to make a long straight wire circuit, the EMF field strength when measured at 0.1m (about 4") from the wire will be about 0.2 milligauss (mG).
NIST, focused on SI units, measures electromagnetic fields in Tesla units of magnetic flux density: magnetic flux density Tesla,
To compute the actual point measurement, each of these numbers, once converted to mG, must be squared, the three squares added, and the square root taken of the sum. This is because the measurement scale is not linear, so a direct raw average would be incorrect.
In the EXCEL worksheet which we provide (at WORKSHEET for EMF MEASUREMENTS) you will observe that provision is made for recording raw data points as well as the individual mG readings. Formulas embedded in the worksheet compute the mean square average in order to obtain a valid point measurement of emf strength by this method.
Please do not contact us with a request buy EMF or RF measuring equipment. We do not sell anything. To do so would be a conflict of interest for this website. These devices are readily available from many electrical equipment and home inspection equipment suppliers. See Evaluation of Low-Cost EMF Instruments This article describes several low-cost and reasonably accurate EMF measurement devices that are readily available. See Radio Frequency RF Detection Meters This article describes several low-cost and accurate radio frequency or RF detection and measurement devices suitable for radio, TV, cellphone, microwave, and similar signals.
Instead of contacting us with a request to perform EMF Electromagnetic or RF Radio Frequency Field Strength measurements, in most cases it is more economical and convenient for a property owner to purchase their own instrument, making measurements under varying conditions. In this series of articles we describe how to make measurements using a consistent approach and using good documentation. See Recommended EMF Measurement Procedure for details of how to collect EMF measurement data.
Following good procedure and using instruments properly are two steps towards making accurate, repeatable EMF measurements. But because the signal transmission for RF sources such as radio, TV, or cell towers, the load on a power transmission line is not under control of an individual property owner, and because the EMF strength varies as the power transmission line load varies, it is important to have an idea of that condition as well when attempting to characterize EMF exposure at a specific location. In contrast, EMF measurements are quite accurate and repeatable at other EMF sources such as close to electrical appliances and service entry cables.
Green link shows where you are in this article series.
Frequently Asked Questions (FAQs)
No FAQs have been posted for this page. Try the search box below or CONTACT US by email if you cannot find the answer you need at InspectApedia.
Try the search box just below or if you prefer, post a question or a comment in the Comments box below and we will respond promptly.
Search the InspectApedia website
HTML Comment Box is loading comments...
Technical Reviewers & References
Related Topics, found near the top of this page suggest articles closely related to this one.