Radio Frequency (RF) Field Measurement Instrument Choices:
This article describes low-cost RF radio frequency electromagnetic field survey measurement instruments useful for performing electromagnetic field RF field detection and level measurements either by engaging a professional or by consumers using low-cost instruments which measure radio frequency level and direction.
We include higher frequency RF measurement instruments such as used for measuring in the marine radar frequency range.
Green links show where you are. © Copyright 2017 InspectApedia.com, All Rights Reserved.
Before buying an EMF or RF test instrument, in order to make sure that the device you are buying is the right one for the hazard you are trying to measure, see EMF RF FIELD & FREQUENCY DEFINITIONS for a simple explanation of different types of radio frequency (RF) and electromagnetic frequency (EMF) types and where they are found.
The information provided here is for research and study purposes. The author makes no representation of unique expertise on this topic, other than having field experience in EMF measurement, having studied technical literature and having conversed with other experts and authors in the field for a number of years.
This information is has not been sanctioned nor technically reviewed by the American Society of Home Inspectors nor the American Industrial Hygiene Association - AIHA. Use it at your own risk.
Radio frequency or RF detection equipment has a more extensive list of applications than the low frequency electromagnetic field EMF detection equipment described separately at Electromagnetic Field EMF ELF Detection Instrument..
While EMF or electromagnetic field low frequency 60 cycle or 60 Hertz signals are associated with power transmission lines and household wiring or appliances, higher frequency RF equipment includes radio and television signals, radio towers, TV towers, cell towers, cellular telephone signals, Wi-Fi signals, bluetooth radio signals, GSM, Radar, DECT, UMTS, WLAN, and all other wireless communication signals.
The cell phone tower in our photo (left) is located in Westchester County, New York. Exposure of electromagnetic radiation from cell towers is normally a separate question from EMR exposure from use of cellphones themselves.
The devices listed just below are for detection and measurement of higher frequency RF or radio frequency detection and measurement. For low frequency EMF (power line type) electromagnetic field detection and measurement, see Evaluation of Low-Cost EMF Instruments.
In these articles we list manufacturers and suppliers of RF and EMF measuring equipment. These devices are readily available from many electrical equipment and home inspection equipment suppliers. Don't contact InspectAPedia to buy equipment - we do not sell anything.
See Evaluation of Low-Cost EMF Instruments This article describes several low-cost and reasonably accurate EMF measurement devices that are readily available suitable for measuring EMF from power transmission lines or home electrical wiring and appliances.
See Radio Frequency RF Detection Meters This article describes several low-cost and accurate radio frequency or RF detection and measurement devices suitable for radio, TV, cellphone, microwave, and similar signals.
B&K Precision offers a variety of instruments to test and characterize RF and microwave devices operating from DC to 8.5 GHz. Here is the company's general product description:
RF Field Strength Meters detect the electric field of radio and microwave signals and measure the electric field strength. Field strength meters provide field technicians and engineers with a cost-effective measurement tool for basic signal investigations at only a fraction of the cost of a full-featured conventional spectrum analyzer.
Handheld RF Field Strength Analyzers are ideal tools for field testing, installing and maintaining Mobile Telecommunications systems such as Cellular and Cordless Phone, CB Paging, Paging, Cable and Satellite TV Systems and performing antenna site measurements and maintenance.
2016/03/07 Marine Radar Units said:
I am a lock and Dam Operator. Recently I was told to start having the pilots of these tow boats put there radars in stand by mode. If they are on, lime they have been for the last 6 years I have been working that they pose a heath hazard to us since we normally work with in 10 foot of these things and are pretty much at eye level with them.
I read the above article in hopes of finding a tool to measure the amount of radiation we are exposed to but didn't find what I was looking for. I may have missed it. Can you help me out. Thanks
This question was posted originally at DEFINITIONS of EMF RF FIELD & FREQUENCY
[Click to enlarge any image] Above, the Bonanza HFW59D_High_Frequency_RF_Meter
Thanks for the questions L&DO:
Radar operates in higher end of "Ultra High Frequency" into the lower end of the "Extremely High Frequency EHF - Microwaves" range given in the table in our article above. That is 30-300 Gigahertz GHz - 109 up to 1012. I'd say in general radar measurement instruments need to measure from 2 GHZ t0 10 GHZ.
From a quick search and quoting from Rice Electronics www.ricelectronics.com, we found this about the radio frequency of marine radar:
There are two basic marine radar frequencies commonly known as "X" and "S" band. "X" band, because of its higher frequency, 10 GHz provides a higher resolution and a crisper image while "S" band, at 3 GHz is less affected by rain and fog.
That means you'd need a measuring instrument that measures field strength in that frequency range. Or a radar receiver. Low frequency "gauss meters" popular at electronics suppliers are not suitable. In fact there are measuring instruments for every RF range, but you may find that in the radar range, say 3 - 10 Gigahertz, they'll be a bit pricey.
Perhaps you can find a source to rent the device, though considering your exposure you might want longer term or even continuous monitoring since otherwise you may not know whose radar is on or off as boats come through your lock.
Special difficulties ensue in making microwave measurements on Navy ships, including cramped space, multiple RF sources, and many sources of RF interference.
Hand-held instruments for shipboard RF measurements are provided by Mobile Design at
We agree with the company's instrument use advice, in particular using the instrument in a consistent, standard, repeatable manner, and taking into account background signals and other sources of interference.
Sources: Zap Checker - http://www.zapchecker.com/ - Alan Broadband, 751 Laurel St #12, San Carlos, CA 94070 USA Ph: 650-345-9627
Safe Living Technologies Inc., 34 Queen Street P.O. Box 72 Morriston, Ontario N0B 2C0 Canada Phone: 519-240-8735 Fax: 519-821-5724
The Zap Checker - a series of hand-held ultra-sensitive instruments intended to detect and display radio frequency (RF) signals over varying spans of radio frequency range. Instrument prices range from around $160 to $2800. (March 2010). The company describes the Zap Checker as a counter surveillance, bug detector, field strength meter. Here is the company's general product line description:
All Zap Checker models are easy to use, small, handheld products with internal fixed antennas which require no adjustments. All models employ linear (regular) as well as logarithmic (compressed) amplification and detection. Each model has an analog meter and LED display. (The LED displays are visible from a distance and at night time.)
Most Zap Checkers have an audio output to hear signal strength levels and identify sources by their characteristic sounds. Some products have a switch enabled silent vibrator. The more professional models employ specialized external antennas and, in the case of the ABC 126, a high frequency band that filters out noise and interfering signals to allow for the highly sensitive detection of signals above 1.5 GHz. The ABC 126 also has a USB output connection for distant monitoring, remote powering and accessory switching.
Instead of contacting us with a request to perform RF Radio Frequency Field Strength measurements, in most cases it is more economical and convenient for a property owner to purchase their own instrument, making measurements under varying conditions. In this series of articles we describe how to make measurements using a consistent approach and using good documentation.
See Recommended EMF Measurement Procedure for details of how to collect RF and / or EMF measurement data.
Following good procedure and using instruments properly are two steps towards making accurate, repeatable RF measurements. But because the signal transmission for RF sources is not under control of an individual property owner, and because the field strength varies as the signal transmission activity varies, it is important to have an idea of that condition as well when attempting to characterize RF exposure at a specific location.
Please do not contact us with a request buy electronic measuring equipment. We do not sell anything. To do so would be a conflict of interest for this website. These devices are readily available from many electrical equipment and home inspection equipment suppliers.
(Jan 8, 2013) Dennis Lee said:
Does the use of two separate 'open door' microwave ovens interference field increase the chances of causing ill health ?
I had a chat with some of my friends and we said yes. What do you think ?
Dennis, the U.S. FDA, back in 1997, issued safety guidance regarding open microwave oven doors - a safety and health hazard that can occur if the microwave oven is not wired properly.
Here is a copy:
US FDA Guidance on Hazards of Open Microwave Oven Doors [PDF] (1997), retrieved 4 June 2015, original source: http://www.fda.gov/downloads/MedicalDevices/DeviceRegulationandGuidance/GuidanceDocuments/UCM095341.pdf
Continue reading at DEFINITIONS of HERTZ, KHz MHz GHz THz or select a topic from closely-related articles below, or see our complete INDEX to RELATED ARTICLES below.
Or see DEFINITIONS of EMF RF FIELD & FREQUENCY if you are not sure which electromagnetic or radio frequency range is of interest to you
Or see these
Articles About Electromagnetic Fields, Hazards, Measurements
- CELL PHONE RADIATION
- DEFINITIONS of EMF RF FIELD & FREQUENCY
- DEFINITIONS of HERTZ, KHz MHz GHz THz
- ELECTROMAGNETIC FIELD EMF ELF & RF DETECTION
- EMF ELECTROMAGNETIC FIELDS & HUMAN EXPOSURE
- EMF CANCER SCARE
- EMF CANCER RISK
- EMF/EMR MEASUREMENT OVERVIEW
- EMF LOCAL SOURCES MAY EXCEED POWER LINE STRENGTH
- EMF MEASUREMENT DISTANCE AFFECTS STRENGTH
- EMF MEASUREMENT INSTRUMENTS
- EMF MEASUREMENT INSTRUMENT ACCURACY
- EMF MEASUREMENT INSTRUMENT USE TIPS
- EMF MEASUREMENT PROCEDURES
- EMF MEASUREMENT STEP BY STEP
- EMF REFERENCES
- EMF SURVEY PROCEDURE
- EMF SURVEY REPORT INTERPRETATION
- EMF WORKSHEET for EMF MEASUREMENTS
- EMF WORKSHEET Example
- EMF WORKPLACE EXPOSURE
- ENVIRONMENTAL HAZARDS - INDOOR - home
- ENVIRO-SCARE - PUBLIC FEAR CYCLES
- RF RADIO FREQUENCY DETECTION METERS
Or use the SEARCH BOX found below to Ask a Question or Search InspectApedia
Try the search box below or CONTACT US by email if you cannot find the answer you need at InspectApedia.
(May 28, 2015)
Thanks for your article, it can help me.
Really your post is really very good and I appreciate it.
It’s hard to sort the good from the bad sometimes,
You write very well which is amazing. I really impressed by your post.
Use the "Click to Show or Hide FAQs" link just above to see recently-posted questions, comments, replies, try the search box just below, or if you prefer, post a question or comment in the Comments box below and we will respond promptly.
Search the InspectApedia website