The purpose of this paper is to help microscopists photograph, store, and use digital microphotographs using modern digital cameras and transmitted-light microscopes.
Discussion focuses on selection of cameras adapt well to microscope eyepieces or trinocular heads, and on camera settings and procedures to obtain best quality photographs. We also discuss image resolution and size settings and make recommendations. Cameras used as examples in this paper include the Nikon Coolpix series 990, 995, and 4500.
[The photograph at page very top is one of our earliest attempts at photographing Aspergillus sp. using lacto phenol cotton blue stain. Even a beginner can obtain very good microphotographs with just a little care. The photograph shown here is of one of our stereoscopic microscopes in use for making digital photographs.
Green links show where you are. © Copyright 2015 InspectApedia.com, All Rights Reserved.
There are many digital cameras on the market that might work to make acceptable photomicrographs. We have found that the Nikon Coolpix series (photo at left) has several advantages for through-eyepiece hand-held photomicrographs: the lens size fits nicely on most eyepieces without scratching anything; the auto focus and other automatic settings produce excellent photographs.
The techniques discussed in this paper work well with digital cameras and any type of microscope, low power stereo zoom to high power forensic polarized light microscopes.]
Digital microphotographs - camera recommendations: here we recommend specific digital cameras that work particularly well with microscopes.
PAAA 2005 Symposium, University of Tulsa, Tulsa Oklahoma - June 2-5, 2005 - updated 02/23/2009
Desirable Features for Digital Camera to Work with Microscopes
We have used the Nikon Coolpix models 990, 995, 4500 (page top photo and photo at left) with quite satisfactory results in preparing more than 750,000 digital photomicrographs and a large number of macro photos as well. The Nikon Coolpix 990, 995, and 4500 will work identically well though some of the menu settings are found in different places.
Camera Features that Do Not Work Well with Microscopes
The Nikon Coolpix 850 (photo at left), and similar models from any manufacturer whose cameras employ external moving telescoping lens components are less successful for photomicrography. The sliding/telescoping barrel does not provide a rigid mount against the microscope eyepiece.
Though it's official macro-lens close-up range is not as close as the other two models, this camera's auto-focus is able to adjust itself for successful photomicrographs. These Nikon models are available as new, factory reconditioned, and used from EBay and from several large camera dealers at prices typically in the $300-$400. range depending on accessories included.
Cameras with an external telescoping lens barrel that moves in and out during focus adjustment. The camera lens focusing mechanism on most digital cameras that use this design is intended to move just the lens barrel and it is not intended to carry the entire weight of the camera.
Even though lens adapters can be purchase to adapt most cameras to a microscope eyepiece, if your camera is going to sit atop an eyepiece and then has to lift the entire camera body up and down as it attempts to auto-focus on the microscope image you are likely to damage the focusing mechanism.
We have also observed that the wobble and slack in movable external lens barrels on digital cameras may prevent good solid mounting and focus when used on a microscope.
Using other camera models made by Nikon, Canon, Kodak, or other manufacturers may be satisfactory, but we have found that the lens diameter and the internal rather than external telescoping parts of this Nikon 900/4500 series works particularly well for photomicrography.
Some of the Canon digital cameras have superb lenses and outstanding macro photography ability, but the large lenses on these cameras do not so perfectly fit the microscope eyepiece and field size. Other smaller hand-held digital cameras
We have used, such as the Canon digital elph can be used with the microscope but results were disappointing as the image formed is very small and telescoped. Microscopists may want to avoid some of the older small-pixel small-format earlier systems which produce only very small, low-resolution images.
Special systems which connect the camera output directly to a computer for on-computer image capture are particularly efficient for in-lab use. we have not (yet) included specifics on these features in this paper as herein we are focusing on image formation at the microscope and in the camera.
Cameras with an appropriate front lens diameter can be hand-held against a microscope eyepiece to take successful photographs. Adapters which screw into the front of the camera lens and slide into a microscope eyepiece tube or into a trinocular head are a better approach for regular use in the laboratory.
We use four such cameras permanently mounted on microscopes in our lab, for example.
My mount supplier also provided a mount stabilizing ring which provided a more secure position on our trinocular stereoscope.
In a comparison test using the same camera hand-held at eyepiece and mounted on the trinocular head with two different adapters we observed no detectable difference in the image produced with eyepiece mounts and hand-held shots.
Digital Camera Features that Work Well with Microscopes
Question: are there newer model digital cameras with more pixels that work as well or better than the Nikon Coolpix 990, 995, and 4500 series?
I have read your interesting guide to selection of digital cameras for microphotography.You recommend the Nikon Coolpix models 990, 995, 4500 for their ease in use. Cameras of those models are still available. Do you know if any of the newer digital models with greater number of pixels are also compatible for microscopic photography. - Fred Colley, Pyxis Laboratories, LLC, Portland, OR
We have followed with interest the continuing improvements in lenses, image chips, and other features in ensuing generations of digital cameras, and many of them produce superior photo images in general use. But for microphotography we have not encountered newer digital camera models with the following features that work particularly well with microscopes compared with the camera series we described above.
Relationship of Camera Lens Physical Size and Microscope Eyepiece - optics & stability
For the models included in the Nikon camera series we have described, the camera's front lens diameter is very close to the diameter of an eyepiece, so with an adapter it is optically efficient, stable, and not too heavy on the microscope. In fact when traveling to another lab, with no mounting adapter at all, if someone has something interesting in focus on their microscope, it takes only an instant to simply hold the camera lens against an eyepiece and snap a photo. This feature has been invaluable when attending conferences on or courses in microscopy when there is a room full of microscopists busily discovering interesting things.
Mechanical Stability and Stress on Camera Moving Parts When Mounted to a Microscope
Unlike other digital cameras whose front lens telescopes "out" from the camera body as focal length changes, all of the moving parts are inside the camera series we recommend above. As a result, when the camera with adapter is mounted onto the microscope it is not only stable, but there is no stress on any moving camera parts as it changes focus. If you zoom out other camera models and (gently) move the lens barrel you will see that those components are not very strong nor are they immune to lateral movement.
More Camera Pixels Does Not Necessarily Mean a Better Photographic Image
More pixels does not necessarily mean sharper images through the microscope. Many consumers are misled in this matter as manufacturers advertise higher pixel numbers for their newer camera models. The resolution ability of the camera lens as well as the design of the image-capturing chip remain the determinants of sharpness of photo images in general.
What more pixels does buy you is the ability to print a larger physical image on paper without loss of resolution, or the ability to zoom in and crop special features in a photograph. The camera series we describe, at the settings we usually use, easily produces high resolution 8" x 10" photographs. If you need to print larger images you will want to set the camera to its higher pixel or higher image size. If you are never printing large images nor cropping onto image features, capturing and storing 4 MB pixel images may be a waste of storage space. (Disk space is increasingly cheap however the time to backup, copy, move libraries still increases with the size of images stored.)
Importance of the Camera's Auto focus in Macro Mode for Microphotography
But more important in capturing sharp images in the microscope in digital photos is the ability of a particular camera model to focus optimally when attached to a microscope - a feather that also varies widely among cameras. The macro focus and auto focus ability of the Nikon series we mention works very well in this regard, so that even when the microscope is not focused perfectly for the camera (you focused it for your eye), the auto focus will often produce an improved, sharp image.
When Do We Set the Camera Focus on Infinity for Microphotography?
Only when the auto focus cannot find an object in the field of view suitable for auto focus do you need to set the camera focal length on infinity and then focus the microscope optimally for the camera by hand, watching the image sharpness on the camera's LCD screen. In this case as well, the Nikon series I recommend offer an added advantage.
Importance of the Camera's Rotating Body and/or Viewing Screen to 90 Deg. From the Lens Axis
Because the camera body can rotate 90 degrees from the axis of the lens itself, you can mount the camera facing down on a microscope eyepiece or on a trinocular head, while rotating the image screen vertically so that its image is easily seen by the operator seated at the microscope workbench.
Other model cameras that do not have a rotating body but that can mount with stability onto the microscope will present their viewing screen in the "up" position on the camera's back. Unless the viewing screen is one that can rotate to a more convenient position, the operator will have to stand up and perhaps even tiptoe to look "down" onto the screen for focusing or image checking.
In sum, as you can see in our photos above of a camera mounted on a microscope, the ability to aim the camera viewing screen comfortably towards a seated microscopist is a useful feature.
Alternative Camera Models for Microphotography
Although there are indeed lens adapters that can mount other digital cameras to a microscope, if you are not going to use one of the models I've described, you will want to go to a much larger, SLR type model that also incorporates a physically strong and stable lens whose moving parts are either internal or are much stronger than the small zooming digital cameras on the market.
Certainly we have no economic interest in recommending a particular piece of equipment, and if you find other camera models that work well in all of the regards I've discussed, I'd be very glad to add that information to our description, particularly because over time the availability of the models we have listed will surely decline.
We welcome any debate, opinion, or suggestion that adds to this information.
Green link shows where you are in this article series.
Frequently Asked Questions (FAQs)
No FAQs have been posted for this page. Try the search box below or CONTACT US by email if you cannot find the answer you need at InspectApedia.
Questions & answers or comments about which digital cameras work best with microscopes
Use the "Click to Show or Hide FAQs" link just above to see recently-posted questions, comments, replies, try the search box just below, or if you prefer, post a question or comment in the Comments box below and we will respond promptly.
Search the InspectApedia website
HTML Comment Box is loading comments...
Technical Reviewers & References